Also, according to the AHA, continuous waveform capnography along with clinical assessment is the most reliable method of confirming and monitoring correct placement of an ET tube.
What is Waveform Capnography
Quantitative waveform capnography is the continuous, noninvasive measurement and graphical display of end-tidal carbon dioxide/ETCO2 (also called PetCO2). Capnography uses a sample chamber/sensor placed for optimum evaluation of expired CO2. The inhaled and exhaled carbon dioxide is graphically displayed as a waveform on the monitor along with its corresponding numerical measurement.
As an assessment tool during CPR, capnography can help the ACLS provider determine a number of things. It is a direct measurement of ventilation in the lungs, and it also indirectly measures metabolism and circulation. For example, a decrease in perfusion (cardiac output) will lower the delivery of carbon dioxide to the lungs. This will cause a decrease in the ETCO2 (end-tidal CO2), and this will be observable on the waveform as well as with the numerical measurement.
Normal ETCO2 in the adult patient should be 35-45 mmHg.
Two very practical uses of waveform capnography in CPR are: 1.) evaluating the effectiveness of chest compressions, and 2.) identification of ROSC. Evaluating the effectiveness of chest compressions is accomplished in the following manner: Measurement of a low ETCO2 value (< 10 mmHg) during CPR in an intubated patient would indicate that the quality of chest compressions needs improvement.
High quality chest compressions are achieved when the ETCO2 value is at least 10-20 mmHg.
When ROSC occurs, There will be a significant increase in the ETCO2. (35-45 mmHg) This increase represents a drastic improvement in blood flow (more CO2 being dumped in the lungs by the circulation) which indicates circulation.
For the intubated patient in cardiac arrest, quantitative waveform capnography is now considered the desired method for monitoring the quality of chest compressions and determining when the patient has a ROSC.